Happy, to share my receptions, for yours. Yes, I agre that you, moved my post, sorry for posting in the wrong place.
28.2E U.K. beams 0-24 hour DX receptions in Hungary II.
I presenting to you, my second testing result, and receptions measurements report, now to the 11306 V frequencies. (The testing date: 2021.12.11)
2. Testing results, details
The firstly, and the nowing testing process difference, that i applicated the minimal changing the anything system devices, and i finished also few correction settings:
- the changing, that i applicated: used me, the new, designed by me, second version scalar feedhorn
- Modified to the LNB skew,
- Rim behind the scalar ring, down to the zero points,
- Dish elevation setting up to, cca. + 1 degrees
The early testing showed, that the 11306 frequency H, and V polarity, not 90 degrees int the received. Therefore, this 11306 V polarisation, new LNB skewing needed ago, this cca. -1 degrees moving, to accessing to the best signal. (That realized true? Don’t know securely, but this is my experience.)
2. Measurement report
The second frequency testing, difficult reception task.
28.2E 2F UK beam 11306 V https://www.lyngsat.com/muxes/Astra-2F_UK_11306-V.html
The SNR, in the 11306 V polarity cases to 7,3 dB maximum (Picture, 11306V SNR Max.jpg). If you remember my first study, or measurements report, this 11306 H in the case of polarity power maximum to 5,6 dB ago.
The upper and lower wave, or valley-period strong fluctuation of the signal level, observed in now. (Picture, 11306V SNR Min.jpg)
(I think maybe the scalar ring phase interference and/or quenching causes.)
Picture 11306V End.jpg shows the closing time of the entire reception interval with 6.8 dB SNR values. Uptime: 24:03 HH:MM, BER:4,7E-3
Conclusion of the second measuring:
The testing also now was successful, this is the receptions also are available daily 0 to 24 hours, its my QTH . (Great! Action, Movies, and GB News HD, etc.)
Thank you for your attention, I welcome all corrections, recommendations, comments, clarifications and additions, and I am also available for further questions.
Thank, you this reading also my second report.
My QTH location:SatQTHmap – Google My MapsKöszönet minden műholdvadásznak, akik megosztják a vételi lehetőségeiket a közösséggel. Feliratkozás: email@example.com
This maps showinng, many more Hungarian DX-r friends also, select and klicking the actual nicknames. The page autozooms in the concrete areas and showing the QTH palce, plus more leftt side, list more DX receptions, and pictures.
Yes, i now using the Invacom QTF-031. This LNB, very good working the cross polarity isolations, this is function very important int the DX, and my LNB testing also showing, that this is LNB the biggest sensitive converter.
I few (one or two) years maked one LNB test, is Invacom, Inverto, MTI, and more with products. If would like, i' am happy to share it , this report/post.
The list says that there are channels, but unfortunately, after searching for this transponder, nothing scans anything, but the signal beam shows as much as 11db.
Skandynavia (near Malmó) 180cm antenna
channels went from that frequency. I think it was a test !!
They are testing 11306 H 27500 again
Stable reception Antenna 180cm Skandynavia (near Malmó)
11306 H 27500 5/6 (copy of 10964 H 22000 5/6)
Alles was vom UK-Beam hier ankommt wird doch immer gern genommen
Ähnlich, aber eine Spur noch weniger wie im Beitrag #11 von Stephan94 kommt im nördlichen Niederösterreich an.
Hätte eigentlich schlechteres erwartet.
dF signal loop must be more, then +/-dF signal search